Shutter Configuration Settings Panel
Shutter Daq Devices and Channels Settings
|Shutter Name||Enter a name for the Shutter.|
Select the DAQ or RIO device for the Shutter.
ScanImage determines the DAQ devices configured on the system on which ScanImage is running.
|Channel ID||Select the channel corresponding to the selected Daq Name for the Shutter.|
|Shutter Open Digital Level||Select either High or Low to indicate the TTL level corresponding to the open state for the Shutter line.|
|Delete Row||To delete a row, press the "X" on the row you want deleted.|
- One default Shutter Name row is provided for a new ScanImage Machine Data File (MDF).
- The default entry is given the name "Main Shutter", using the PXI1Slot3, on the PFI12 Channel with a "High" digital level.
- This row can be deleted.
- All columns can be edited.
Bottom Panel Display Settings
|Shutter Transition Time (seconds)||Enter the time, in seconds, to delay following certain shutter open commands (e.g. between stack slices), allowing shutter to fully open before proceeding.|
A default value of 0.1 is provided.
image2016-6-8 11:40:23.png (image/png)